The measurement of eyelid margin pressure on the ocular surface [Abstract]
Shaw, Alyra J., Collins, Michael J., Davis, Brett A., & Carney, Leo G. (2009) The measurement of eyelid margin pressure on the ocular surface [Abstract]. In The Association for Research in Vision and Ophthalmology 2009 Annual Meeting, May 3 - 7, Fort Lauderdale, Florida, US.
Purpose:To estimate the pressure of the eyelid against the ocular surface we developed a novel method using a thin (0.18 mm) tactile piezoresistive pressure sensor (Tekscan Inc., Boston, MA, USA) mounted on a rigid contact lens.------
Methods:Curvature of the sensor causes an offset and altered sensitivity, so a contact lens was designed with a flat region where the sensor could be adhered. We examined the response of the sensor in different conditions including the influence of conditioning (prestressing), drift (continued increasing response with a static load) and temperature variations. Techniques were developed to calibrate the sensor output into real pressure units (mmHg) and to hold and optimally place the sensor-contact lens combination under the eyelid margin.------
Results:Calibration gave a linear relationship between raw sensor output and actual pressure units, for loads between 1 and 10 mmHg. Conditioning the sensor prior to use regulated the measurement response. Sensor output stabilised about 10 seconds after loading. While the output drifted slightly over several hours, the drift was not significant over the measurement time required for eyelid pressure (<30 seconds). The error associated with calibrating at room temperature but measuring at ocular surface temperature led to a very small overestimation of pressure. Eyelid pressure readings could be seen when the upper eyelid was placed on the sensor and then removed three times during a recording. Also when the eyelid was pulled from the outer canthus (similar to the lid-pull technique for removing a contact lens) the readings from the sensor significantly increased.------
Conclusions:A number of novel procedures were developed to use a tactile sensor to measure eyelid pressure. These included designing a custom contact lens with a flat area to which the sensor could be attached, a hydrostatic calibration system and an in vivo measurement apparatus to hold and position the sensor-contact lens combination on the eye.
Impact and interest:
Citation counts are sourced monthly from and citation databases.
These databases contain citations from different subsets of available publications and different time periods and thus the citation count from each is usually different. Some works are not in either database and no count is displayed. Scopus includes citations from articles published in 1996 onwards, and Web of Science® generally from 1980 onwards.
Citations counts from theindexing service can be viewed at the linked Google Scholar™ search.
Full-text downloads displays the total number of times this work’s files (e.g., a PDF) have been downloaded from QUT ePrints as well as the number of downloads in the previous 365 days. The count includes downloads for all files if a work has more than one.
|Item Type:||Conference Item (Presentation)|
|Keywords:||eyelid, contact lens|
|Subjects:||Australian and New Zealand Standard Research Classification > MEDICAL AND HEALTH SCIENCES (110000) > OPTOMETRY AND OPHTHALMOLOGY (111300)
Australian and New Zealand Standard Research Classification > MEDICAL AND HEALTH SCIENCES (110000) > OPTOMETRY AND OPHTHALMOLOGY (111300) > Optometry and Ophthalmology not elsewhere classified (111399)
|Divisions:||Current > QUT Faculties and Divisions > Faculty of Health
Current > Institutes > Institute of Health and Biomedical Innovation
Current > Schools > School of Optometry & Vision Science
|Copyright Owner:||Copyright 2009 [please consult the authors]|
|Deposited On:||19 May 2009 23:58|
|Last Modified:||09 Jun 2010 13:40|
Repository Staff Only: item control page