Power profiles of multifocal contact lenses and their interpretation
Plainis, Sotiris, Atchison, David A., & Charman, W. Neil (2013) Power profiles of multifocal contact lenses and their interpretation. Optometry and Vision Science, 90(10), pp. 1066-1077.
Many contact lens (CL) manufacturers produce simultaneous-image lenses in which power varies either smoothly or discontinuously with zonal radius. We present in vitro measurements of some recent CLs and discuss how power profiles might be approximated in terms of nominal distance corrections, near additions, and on-eye visual performance.
Fully hydrated soft, simultaneous-image CLs from four manufacturers (Air Optix AQUA, Alcon; PureVision multifocal, Bausch & Lomb; Acuvue OASYS for Presbyopia, Vistakon; Biofinity multifocal- ‘‘D’’ design, Cooper Vision) were measured with a Phase focus Lens Profiler (Phase Focus Ltd., Sheffield,UK) in a wet cell and powerswere corrected to powers in air. All lenses had zero labeled power for distance.
Sagittal power profiles revealed that the ‘‘low’’ add PureVision and Air Optix lenses exhibit smooth (parabolic) profiles, corresponding to negative spherical aberration. The ‘‘mid’’ and ‘‘high’’ add PureVision and Air Optix lenses have biaspheric designs, leading to different rates of power change for the central and peripheral portions. All OASYS lenses display a series of concentric zones, separated by abrupt discontinuities; individual profiles can be constrained between two parabolically decreasing curves, each giving a valid description of the power changes over alternate annular zones. Biofinity lenses have constant power over the central circular region of radius 1.5 mm, followed by an annular zone where the power increases approximately linearly, the gradient increasing with the add power, and finally an outer zone showing a slow, linear increase in power with a gradient being almost independent of the add power.
The variation in power across the simultaneous-image lenses produces enhanced depth of focus. The throughfocusnature of the image, which influences the ‘‘best focus’’ (distance correction) and the reading addition, will vary with several factors, including lens centration, the wearer’s pupil diameter, and ocular aberrations, particularly spherical aberration; visual performance with some designs may show greater sensitivity to these factors.
Impact and interest:
Citation counts are sourced monthly from and citation databases.
These databases contain citations from different subsets of available publications and different time periods and thus the citation count from each is usually different. Some works are not in either database and no count is displayed. Scopus includes citations from articles published in 1996 onwards, and Web of Science® generally from 1980 onwards.
Citations counts from theindexing service can be viewed at the linked Google Scholar™ search.
|Item Type:||Journal Article|
|Keywords:||contact lenses, multifocal, addition, spherical aberration, pupil, presbyopia|
|Subjects:||Australian and New Zealand Standard Research Classification > MEDICAL AND HEALTH SCIENCES (110000) > OPTOMETRY AND OPHTHALMOLOGY (111300) > Vision Science (111303)|
|Divisions:||Current > QUT Faculties and Divisions > Faculty of Health
Current > Institutes > Institute of Health and Biomedical Innovation
Current > Schools > School of Optometry & Vision Science
|Copyright Owner:||Copyright 2013 American Academy of Optometry|
|Deposited On:||19 Jan 2014 23:18|
|Last Modified:||22 Jan 2014 03:48|
Repository Staff Only: item control page