Biomedical Study of Top Screw Pullout in Anterior Scoliosis Instrumentation
Mayo, Andrew, Labrom, Robert D., Askin, Geoffrey N., & Adam, Clayton J. (2007) Biomedical Study of Top Screw Pullout in Anterior Scoliosis Instrumentation. In Williamson, Owen (Ed.) Annual Scientific Meeting of the Spine Society of Australia, 20-22 April, 2007, Hobart, Australia.
ABSTRACT ID 139
BIOMECHANICAL STUDY OF TOP SCREW PULLOUT IN ANTERIOR SCOLIOSIS CORRECTION
INTRODUCTION Failure of the top screw in anterior scoliosis correction is a common problem with rates of 5-15% reported in the literature. Previous biomechanical studies of anterior vertebral body screws have evaluated their resistance to either straight pullout or cephalo-caudad compression forces with different definitions of failure. The aim of this study was to assess screw resistance to more realistic loading conditions, namely cephalo-caudad compression between adjacent screw heads, followed by a pullout force directed in an arc centred on the screw below with the construct’s rod as it's radius. The effects of degree of compression, screw angulation, and pullout arc radius on vertebral body screw pullout strength were investigated.
METHOD Four tests were performed using sawbone and one with sheep vertebrae. Firstly, straight pullout tests of screws inserted at different angles were performed to define optimum angle. Secondly, a test rig that allowed the screws to be pulled out in an arc of four different radii tested screws inserted straight into the sawbone. Thirdly, the rig was used to test screws inserted at different angles (from 20 caudad to 20 cephalad in steps of 10). Fourthly, a surgeon performed 3 compressions of screws inserted straight into a strain gauged model to assess maximum compression force. Screws inserted into sawbone with or without a staple then had the maximum force achieved by the spinal surgeon applied to them prior to pullout testing. Finally an "ideal" configuration for the screws was decided, and 29 sheep vertebral bodies were instrumented either in the traditional or new "ideal" way to compare pullout strength between the two groups. Results were analysed using pooled t-tests.
RESULTS Sawbone testing revealed that increasing the radius of pullout increased the pullout strength. With a radius of 23mm (the average distance between a T5 and T6 vertebral body screw) a mean maximum pullout force of 392 N was required. For a radius of 33mm (the maximum achievable distance between a T5 and T6 vertebral body screw in an average sized person) the mean maximum pullout force increased to 578N (n=3, P<0.01). An angle of 10 in a cephalad direction was the strongest (at a radius of 33mm the mean maximum pullout of a 10 cephalad screw was 615N compared with 531N at 20 cephalad (n=3, P<0.01) and 578N at 0 (n=3, P<0.05).
The maximum compression force achieved by the spinal surgeon was just less than 800N. Non-compressed screws were strongest and those compressed without staples weakest. Sheep vertebrae instrumented in a traditional manner (23mm between screws inserted parallel to the endplates n=15) achieved pullout = 993N compared with our "ideal" configuration (33mm between screws inserted at 10 cephalad angle to the endplate n=14) 1864N (P<0.01). The screw lengths and bone densities of the vertebrae were measured with no significant difference between groups.
DISCUSSION The most important factor in pullout of the top anterior vertebral body screws is the distance between the top two screws which can be maximized by placing the top screw as close as possible to the top endplate and the bottom screw as close as possible to the bottom endplate (although this will have detrimental effects on the pullout of the second screw should the top screw pull out). Screw angulation is a less important factor but any angulation should be in a cephalad direction and around 10 in magnitude. Our results also suggest that the use of a staple may play a small role in resistance of cephalo-caudad compression.
Impact and interest:
Citation countsare sourced monthly fromand citation databases.
These databases contain citations from different subsets of available publications and different time periods and thus the citation count from each is usually different. Some works are not in either database and no count is displayed. Scopus includes citations from articles published in 1996 onwards, and Web of Science® generally from 1980 onwards.
Citations counts from theindexing service can be viewed at the linked Google Scholar™ search.
|Item Type:||Conference Paper|
|Subjects:||Australian and New Zealand Standard Research Classification > ENGINEERING (090000) > BIOMEDICAL ENGINEERING (090300) > Biomechanical Engineering (090302)|
|Divisions:||Past > QUT Faculties & Divisions > Faculty of Built Environment and Engineering|
Current > Institutes > Institute of Health and Biomedical Innovation
|Deposited On:||09 May 2007|
|Last Modified:||11 Aug 2011 01:22|
Available Versions of this Item
- Biomedical Study of Top Screw Pullout in Anterior Scoliosis Instrumentation. (deposited 09 May 2007)[Currently Displayed]
Repository Staff Only: item control page